Welcome to our site! EDAboard.com is an international Electronic Discussion Forum focused on EDA software, circuits, schematics, books, theory, papers, asic, pld, 8051, DSP, Network, RF, Analog Design, PCB, Service Manuals... and a whole lot more! To participate you need to register. Registration is free. Click here to register now.
C2 and C3 will be dominated by the pads of your test fixture - normally these can be deembedded if calibrated in TRL. If not you can estimate them from the C=epsilon*surface/thickness formula. Then what remains is C1, this can be extracted from the zero in your S21 measurement. This simple model will be adequate for most applications.