+ Post New Thread
Results 1 to 12 of 12

6th January 2013, 15:05 #1
 Join Date
 Jan 2012
 Location
 Italy
 Posts
 576
 Helped
 39 / 39
 Points
 3,048
 Level
 12
Opamp AC characteristics measurement
Dear Friends
I finished the simulation of my opamp design. Now I would like to know the real measurement setup for every test. Starting here in thispost with simulating the AC characteristics (Open loop gain, f3dB, Phase Margin) .
As you know that dealing with an open loop amp is really difficult, therefore an approach like the one I attached is used for this purpose. The problem I dont understand how it working and by which equiepemnt we can scan the frequency and how we can read it to be like the one we simulate.
Please i really need your kind help
Regards

6th January 2013, 15:05

6th January 2013, 16:39 #2
 Join Date
 May 2008
 Location
 Germany
 Posts
 5,642
 Helped
 1692 / 1692
 Points
 39,256
 Level
 48
Re: Opamp AC characteristics measurement
Hi Junus,
the shown test setup has unity gain for dc only  that is necessary to ensure an operating point within the opamp´s linear range. Otherwise, the offset voltage would bring the output into saturation because of the large openloop dc gain.
The capacitor C should be as large as possible because  together with the large R  it creates a large time constant equivalent to a very low corner frequency wc.
Thus, for frequencies above wc the inverting input (increasingly) is ac grounded and you can measure/simulate the openloop gain by connecting a swept frequency signal at Vin.
The load componenets (RL and CL) are not too important. Thus, some kind of normal operation conditions are established..
Remark: The rising linear curve starting at 1/RC is the inverse feedback factor 1/Hf with Hf=1/(1+sRC). Thus, the bold line is identical to the gain of the whole circuit (with feedback).
The crossing of both functions gives you the frequency w1=Av(0)/RC above which the measurement is identical to the desired openloop gain of the opamp alone. This curve clearly shows why you should select a large time constant RC.Last edited by LvW; 6th January 2013 at 16:55.
1 members found this post helpful.

6th January 2013, 19:25 #3
 Join Date
 Jan 2008
 Location
 Bochum, Germany
 Posts
 39,140
 Helped
 11962 / 11962
 Points
 227,185
 Level
 100
Re: Opamp AC characteristics measurement
by which equiepemnt we can scan the frequency and how we can read it to be like the one we simulate
1 members found this post helpful.

6th January 2013, 19:25

6th January 2013, 19:29 #4
 Join Date
 Jan 2012
 Location
 Italy
 Posts
 576
 Helped
 39 / 39
 Points
 3,048
 Level
 12
Re: Opamp AC characteristics measurement
Thank you very much LVW
your explanation is very sufficient , I think you are a teacher in this subject really
I hope to see your replies with the next measurements I would post
Thank you in advance Sir
Regards
   Updated   
Kindly, If it is possible for you to suggest me the equipment,
Any way how could I measure the phase Margin ??

6th January 2013, 21:30 #5
 Join Date
 May 2008
 Location
 Germany
 Posts
 5,642
 Helped
 1692 / 1692
 Points
 39,256
 Level
 48
Re: Opamp AC characteristics measurement

6th January 2013, 21:30

7th January 2013, 00:52 #6
 Join Date
 Jan 2012
 Location
 Italy
 Posts
 576
 Helped
 39 / 39
 Points
 3,048
 Level
 12

7th January 2013, 07:56 #7
 Join Date
 Jan 2008
 Location
 Bochum, Germany
 Posts
 39,140
 Helped
 11962 / 11962
 Points
 227,185
 Level
 100
Re: Opamp AC characteristics measurement
You don't primarly measure phase margin, you determine the phase of the open loop transfer function which allows to calculate phase margin for specific feedback factors.
The phase can be measured with an oscilloscope if you don't have special instruments for this purpose (e.g. a vector network analyzer). The input signal or a respective reference signal from the signal generator have to be connected to the oscilloscope too. Modern DSOs have suitable measurement functions to determine phase differences accurately.
1 members found this post helpful.

7th January 2013, 08:06 #8
 Join Date
 May 2008
 Location
 Germany
 Posts
 5,642
 Helped
 1692 / 1692
 Points
 39,256
 Level
 48
Re: Opamp AC characteristics measurement
As an alternative, you can measure the step response of the closedloop using the oscilloscope. The overshoot you can observe is a good indication of the phase margin.
There are graphics overshoot=f(phase margin), which can be used for this purpose.
This relationship normally holds for second order circuits only, but can be used with good accuracy also for thirdorder systems (this sufficient for opamp applications).
1 members found this post helpful.

7th January 2013, 08:06

7th January 2013, 15:15 #9
 Join Date
 Jan 2012
 Location
 Italy
 Posts
 576
 Helped
 39 / 39
 Points
 3,048
 Level
 12
Re: Opamp AC characteristics measurement
Thank you very much guys for your kind explanation, I will think more about it to ask you further
Have fun
Best Regards

8th January 2013, 00:57 #10
 Join Date
 Jan 2012
 Location
 Italy
 Posts
 576
 Helped
 39 / 39
 Points
 3,048
 Level
 12
Re: Opamp AC characteristics measurement
Dear FvM and LvW
I have found this paper to calculate the phase margin from the step response overshoot , but the equation he used is not clear. Kindly, do you have any clear article regarding the same methods ??
many thanks

8th January 2013, 12:10 #11
 Join Date
 May 2008
 Location
 Germany
 Posts
 5,642
 Helped
 1692 / 1692
 Points
 39,256
 Level
 48
Re: Opamp AC characteristics measurement
Junus,
I think  the equation is quite clear: PM as a function of the damping factor, which is identical to 1/2Q.
However, this requires the knowledge of Q for the closed feedback loop.
Therefore, I attach a copy from a textbook (S. Franco: Design with opamps and analog int. circuits).
There you can find a graph overshoot=f(PM).
1 members found this post helpful.

8th January 2013, 14:14 #12
 Join Date
 Jan 2012
 Location
 Italy
 Posts
 576
 Helped
 39 / 39
 Points
 3,048
 Level
 12
Re: Opamp AC characteristics measurement
Thank you very much LvW, so nice
+ Post New Thread
Please login