+ Post New Thread
Results 1 to 8 of 8

2nd June 2019, 20:49 #1
 Join Date
 Feb 2019
 Posts
 18
 Helped
 0 / 0
 Points
 197
 Level
 2
differential probe with oscilloscope
Hi,
I want to measure the pole voltages for three phases of inverter b6 bridge.
Can anyone tell me shortly how to measure the pole voltage of R phase using differential probe connection to oscilloscope to see square waveform (switchings) of R phase?
in this case do i need two differential probes? since we have two gates i.e., one pole has two mosfet's.
Thanks

Advertisement

2nd June 2019, 21:13 #2
Awards:
 Join Date
 Apr 2014
 Posts
 15,607
 Helped
 3556 / 3556
 Points
 77,026
 Level
 67
Re: differential probe with oscilloscope
Hi,
Honestly, I don't understand the question.
Differential voltage...measures the voltage between two points. Let's name them point A and point B.
A differential voltage probe has two connections for measurement. Connect them to A and B respectively.
KlausPlease don´t contact me via PM, because there is no time to respond to them. No friend requests. Thank you.

2nd June 2019, 22:52 #3
 Join Date
 Jan 2008
 Location
 Bochum, Germany
 Posts
 45,250
 Helped
 13765 / 13765
 Points
 259,183
 Level
 100
Re: differential probe with oscilloscope
In practice you'll measure pole voltage either referred to DC bus minus or plus.

Advertisement

3rd June 2019, 06:09 #4
 Join Date
 Apr 2011
 Location
 Minneapolis, Minnesota, USA
 Posts
 12,875
 Helped
 2568 / 2568
 Points
 52,584
 Level
 56
Re: differential probe with oscilloscope
I presume this resembles your setup. My simulation is 3phase delta. Each phase created from 2 transistors. The gating signals are square waves. Shootthrough does not occur. Waveform to loads is called modified sine waves, although they are square waves with brief dead time.
How many channels on your oscilloscope? If 2 channels then you can attach the ground clip to any of the load nodes, and read two load waveforms directly. To figure the third load, however, you need to do math in your head. (This method creates uncertainty about displaying current directions in proper relation.)
If you wish to examine current directions, then you need 3 differential scope probes. A simulator makes it easy but real hardware needs real oscilloscopes.
So as a handy trick, make 2 antiparallel led's as each load. Reduce frequency to 1 or 2 Hz. This makes it easier to watch the led's indicate current direction and timing.

3rd June 2019, 11:08 #5
 Join Date
 Feb 2019
 Posts
 18
 Helped
 0 / 0
 Points
 197
 Level
 2
Re: differential probe with oscilloscope
Hi BradtheRad,
I want to measure switching waveform at one phase(i.e., for two mosfet’s). In this case do i need two differential probes for connecting at two gates and two sources of two mosfets?

Advertisement

3rd June 2019, 11:28 #6
 Join Date
 Jan 2008
 Location
 Bochum, Germany
 Posts
 45,250
 Helped
 13765 / 13765
 Points
 259,183
 Level
 100
Re: differential probe with oscilloscope
You are asking about "pole" voltage. Why do you want to measure the gate voltages? They can be of course interesting, but in this case I count three probes: low and highside Vgs + lowside Vds, if Vdc can be considered constant, thus highside Vds has same waveform as lowside.

3rd June 2019, 12:54 #7
 Join Date
 Feb 2019
 Posts
 18
 Helped
 0 / 0
 Points
 197
 Level
 2

3rd June 2019, 15:42 #8
 Join Date
 Jan 2008
 Location
 Bochum, Germany
 Posts
 45,250
 Helped
 13765 / 13765
 Points
 259,183
 Level
 100
Re: differential probe with oscilloscope
Vds,hs = Vdc  Vds,ls (if measured drain to source in both cases). Thus it's inverted and DC shifted, but basically the same AC waveform, as long Vdc is constant. In a real bridge inverter, Vdc may carry some ripple, making Vds,hs looking different from Vds,ls. But that's beyond the original question of "measure the pole voltages".
+ Post New Thread
Please login