+ Post New Thread
Results 1 to 7 of 7

24th November 2016, 14:34 #1
 Join Date
 Mar 2014
 Posts
 60
 Helped
 0 / 0
 Points
 887
 Level
 6
Noise in AC voltage measurement
Hi,
I am trying to measure AC rms voltage using PIC microcontroller. I have attached the circuit for measurement. I am connecting a 3V step down transformer out to the input of the circuit.The feedback capacitor has been added to reduce the noise, but I am getting noise if I add this. This is getting smooth after the low pass filtering. Everything else, like clamping the voltage and all are properly happening. Do I really need to place the feedback capacitor? What are the recommended values if it is needed?
Thanks

24th November 2016, 14:34

24th November 2016, 15:30 #2
Awards:
 Join Date
 Apr 2014
 Posts
 10,674
 Helped
 2530 / 2530
 Points
 53,016
 Level
 56
Re: Noise in AC voltage measurement
Hi,
I assume you modified the original circuit...
and now there are some problematic modifications.
according
* bias voltage
* common mode input voltage
* input impedance
* input votage range
* output voltage range
* frequency
****
First of all: You could do completely without OPAMP.
Like this.
***
I see you really want to calculate RMS. This is good. Especially when the waveform differs from sine shape.
On the other hand you attenuate the frequencies above 50Hz... so you get an almost sine shape.
Is this what you want?
Klaus

24th November 2016, 15:30

25th November 2016, 05:55 #3
 Join Date
 Mar 2014
 Posts
 60
 Helped
 0 / 0
 Points
 887
 Level
 6
Re: Noise in AC voltage measurement
On the other hand you attenuate the frequencies above 50Hz... so you get an almost sine shape.
Is this what you want?
thanks

25th November 2016, 05:55

25th November 2016, 07:43 #4
Awards:
 Join Date
 Apr 2014
 Posts
 10,674
 Helped
 2530 / 2530
 Points
 53,016
 Level
 56
Re: Noise in AC voltage measurement
Hi,
C7 is not creating the noise. But it may cause to amplify it.
I assume it is common mode noise. Usually a difference amplifier attenuates common mode noise.
But only if R1 value = R15 value, which isn't the case here (here I assume someone didn't understand the circuit, but modifed it).
Adding C7 to R15 makes this even worse.
In your case you may disconnect C7. To further reduce common mode noise I recommend to use two 100nF capacitors.
Connect them at each side of C4 to GND.
Because this adds another 50nF to C4 and fc is close to mains frequency you may expect up to 10% less output voltage.
If you don't want this you may use 10nF capacitors
****
By far the better solution is to correct the wrong R1 and R15 values. This brings improvement in almost any function of the circuit: noise, common mode input voltage range, symmetry, differential input voltage range...
Klaus

25th November 2016, 09:37 #5
 Join Date
 Mar 2014
 Posts
 60
 Helped
 0 / 0
 Points
 887
 Level
 6
Re: Noise in AC voltage measurement
Hi, thanks for the quick and valuable reply. I'll change the resistors and update the result.

30th November 2016, 06:47 #6
 Join Date
 Mar 2014
 Posts
 60
 Helped
 0 / 0
 Points
 887
 Level
 6
Re: Noise in AC voltage measurement
Hi,
Noise has reduced to a greater extent. I could measure the RMS value with our laboratory setup. I assume there will be much more noise when it is installed in the field. So I should eliminate as much noise I can with this. How to select C1 and C7 values to further reduce noise? Shall I use 100nF in both places?

30th November 2016, 06:47

30th November 2016, 08:12 #7
Awards:
 Join Date
 Apr 2014
 Posts
 10,674
 Helped
 2530 / 2530
 Points
 53,016
 Level
 56
Re: Noise in AC voltage measurement
Hi,
Please give the complete new circuit.
Generally.
C7 with R15 act as a low pass filter.
The same values you need to install at R1 and a new C. ( R1 = R15, new C = C7).
Fc = 1 / ( 2 × Pi × R × C) .... as usual.
Klaus
+ Post New Thread
Please login